To develop a high performance biomedical dry electrode, the laser micromilling-recasting technology is used to fabricate the metal dry electrode with surface micostructure arrays. Based on the analysis of the micro morphology of the electrode surface, the wettability of the electrode surface is discussed, and then the influence of laser processing parameters such as scanning spacing, scanning speed and scan times on the adhesion performance of Escherichia coli is further investigated. The results show that the contact angle of metal dry electrode with surface microstructure arrays fabricated with reasonable laser processing parameter can reach more than 150° and show the superhydrophobic characteristics. The adhesion performance of escherichia coli of electrode is changed greatly with different scanning spacing and scan times. When the 0.1 mm scanning spacing is selected, the least amount of escherichia coli is found on the surface of electrode. With the increasing scan times, the adhesion amount of escherichia coli can be reduced. However, the scanned speed has little effect on the adhesion performance of escherichia coli for metal dry electrode.
Fabrication and bacterial adhesion of metal dry electrode with surface microstructure arrays
First published at:Dec 15, 2017
Opto-Electronic Engineering Vol. 44, Issue 12, pp. 1187 - 1193 (2017) DOI:10.3969/j.issn.1003-501X.2017.12.006
1 Srivastava A K，Bhartia B，Mukhopadhyay K，et al. Long term biopotential recording by body conformable photolithography fabricated low cost polymeric microneedle arrays[J]. Sensor And Actuators A :Physical，2015，236(9)：164–172.
2 Zhou Wei，Liu Wei，Qiu Qingfu，et al. Development，fabrication，and applications of biomedical electrodes[J]. Chinese Science Bulletin，2015，60(15)：1352–1360.
周伟，刘伟，邱清富，等. 生物医用电极制造技术及应用研究进展[J]. 科学通报，2015，60(15)：1352–1360.
3 Shi Hengchong, Yin Jinghua. Surface construction of antibacterial property on biomedical polymer materials and its application on medical device[J]. Polymer Bulletin, 2016(9): 196–202.
石恒冲, 殷敬华. 医用高分子材料抗菌表面构建及在医疗器械中应用[J]. 高分子通报, 2016(9): 196–202.
4 Wang Zhangming, Sun Wei, Tian Xiubo, et al. Progress in antibacterial field by surface modification[J]. Journal of Mate-rials Science & Engineering, 2006, 24(5): 787–791.
王章明, 孙伟, 田修波, 等. 抗菌表面处理技术研究进展[J]. 材料科学与工程学报, 2006, 24(5): 787–791.
5 Yuan Yue, Hays M P, Hardwidge P R, et al. Surface characteristics influencing bacterial adhesion to polymeric substrates[J]. RSC Advances, 2017, 7(23): 14254–14261.
6 Jiang You, Yin Yijie, Zha Xincheng, et al. Wettability regulated gram-negative bacterial adhesion on biomimetic hierarchical structures[J]. Chinese Chemical Letters, 2017, 28(4): 813–817.
7 Hou Shuyu, Gu Huan, Smith C, et al. Microtopographic patterns affect Escherichia coli biofilm formation on poly(dimethylsiloxane) surfaces[J]. Langmuir, 2011, 27(6): 2686–2691.
8 Zhou Rui, Li Fengping. Synthesis of nanoparticles by short pulsed laser ablation and its applications in nonlinear optics[J]. Opto-Electronic Engineering, 2017, 44(2): 172–184.
周锐，李峰平. 纳米颗粒的短脉冲激光烧蚀制备及其非线性光学应用[J]. 光电工程, 2017, 44(2): 172–184.
9 Yan Huangping, Abdul Rashid M R B, Khew Si Ying, et al. Realization of laser textured brass surface via temperature tuning for surface wettability transition[J]. Opto-Electronic Engineering, 2017, 44(6): 587–592.
10 Guan Y C, Zhou W, Li Z L, et al. Femtosecond laser-induced ripple structures on magnesium[J]. Applied Physics A, 2014, 115(1): 13–18.
11 Lee S W, Shin H S, Chu C N. Fabrication of micro-pin array with high aspect ratio on stainless steel using nanosecond laser beam machining[J]. Applied Surface Science, 2013, 264: 653–663.
12 Zhou Wei, Song Rong, Pan Xiaoling, et al. Fabrication and impedance measurement of novel metal dry bioelectrode[J]. Sensors and Actuators A: Physical, 2013, 201: 127–133.
13 Wang Xiao, Li Liyin, Shen Zongbao, et al. Experimental investigation on: Laser shock micro-forming process using the mask and flexible pad[J]. Optics and Lasers in Engineering, 2017, 88: 102–110.
14 Miyagi M, Zhang Xuedong, Kawahito Y, et al. Surface void suppression for pure copper by high-speed laser scanner welding[J]. Journal of Materials Processing Technology, 2017, 240: 52–59.
15 Zhou Wei, Liu Wei, Liu Shaoyu, et al. Experimental investigation on surface wettability of copper-based dry bioelectrodes[J]. Sensors and Actuators A: Physical, 2016, 244: 237–242.
16 Tang M, Shim V, Pan Z Y, et al. Laser ablation of metal substrates for super-hydrophobic effect[J]. Journal of Laser Micro/Nanoengineering, 2011, 6(1): 6–9.
Get Citation: Liu Shaoyu, Zhou Wei, Li Yaoyao, et al. Fabrication and bacterial adhesion of metal dry electrode with surface microstructure arrays[J]. Opto-Electronic Engineering, 2017, 44(12): 1187–1193.
Previous: Microtube fabrication based on femtosecond Bessel beam and its flexible driving with external magnetic field
Next: Effect of constant temperature substrate on microstructure and hardness of Al2O3-based eutectic ceramics