With the widespread application of ultraviolet spectroscopy, low-cost portable ultraviolet spectrometer has become a research focus in this field. Firstly, the optical structure of the portable UV-VIS spectrometer was de-signed based on the crossed-asymmetric Czerny-Turner structure in the paper. Secondly, the key devices of ultraviolet spectrometer, namely ultraviolet detectors and blazed gratings, were studied. The coated UV-enhanced CCDs were fabricated using Lumogen fluorescent materials and vacuum coating methods. The influence of the position on the CCD surface of the fluorescent film on the resolution was analyzed. The effect of blazed gratings on the multi-order diffraction efficiency in the ultraviolet region was theoretically studied. Finally, the test results of performance of a portable UV-VIS spectrometer prototype show that the resolution of the 200 nm~900 nm band, 25 μm slit width, 600 lp/mm, 300 nm blazed grating configuration is less than 1.5 nm and the spectral responsivity increases to 20% in the spectral range varying from 200 nm to 300 nm, which meets the design requirements of the portable UV-VIS spectrometer.
Design and key technology research of portable UV-VIS spectrometer
First published at:Oct 12, 2018
 Yang J. The application and development of UV detection technology[J]. Optoelectronic Technology, 2011, 31(4): 274–278.
杨杰. 紫外探测技术的应用与进展[J]. 光电子技术, 2011, 31(4): 274–278.
 Weiser H, Vitz R C, Moos H W, et al. Sensitive far uv spectrograph with a multispectral element microchannel plate detector for rocket-borne astronomy[J]. Applied Optics, 1976, 15(12): 3123–3130.
 Chen X, Li Z X, Yan F, et al. Test system for absolute spectral response of SBUV image intensifier[J]. Opto-Electronic Engineering, 2016, 43(5): 8–14.
陈雪, 李宗轩, 闫丰, 等. 日盲紫外像增强器绝对光谱响应测试
系统[J]. 光电工程, 2016, 43(5): 8–14.
 Feng F, Duan F J, Bo E, et al. An optical design of small-size echelle spectrograph[J]. Opto-Electronic Engineering, 2014, 41(7): 20–25.
冯帆, 段发阶, 伯恩, 等. 一种小型中阶梯光栅光谱仪的光学设计[J]. 光电工程, 2014, 41(7): 20–25.
 Franks W A R, Kiik M J, Nathan A. UV-responsive CCD image sensors with enhanced inorganic phosphor coatings[J]. IEEE Transactions on Electron Devices, 2003, 50(2): 352–358.
 Liu Q, Ma S B, Qian X C, et al. Thickness optimization and photoelectric performance test of UV sensitized film of COMS sensor[J]. Acta Photonica Sinica, 2017, 46(6): 0604002.
刘琼, 马守宝, 钱晓晨, 等. CMOS传感器紫外敏化膜层的厚度优化及其光电性能测试[J]. 光子学报, 2017, 46(6): 0604002.
 Alexander S J. Phosphor coated UV-responsive CCD image sensors[D]. Waterloo, Ontario: University of Waterloo, 2002.
 Franks W A R. Inorganic phosphor coatings for ultraviolet responsive image detectors[D]. Waterloo, Ontario: University of Waterloo, 2000.
 Tong J P, Gao J X, Wang F, et al. Nonlinear correction of the sensor S11639 in mini-spectrometer[J]. Opto-Electronic Engineering, 2017, 44(11): 1101–1106.
童建平, 高建勋, 汪飞, 等. 微型光谱仪中传感器S11639的非线性校正[J]. 光电工程, 2017, 44(11): 1101–1106.
 Cheng L, Chen Y P, Zhu R B , et al. Design and Realization of Micro Fiber Spectrometers for Bioluminescence Detecting Systems' Stray Light Detection[J]. Bioorganic & Medicinal Chemistry Letters, 2006, 14(2):427–429.
 Xia G, Wu S, Wang G D, et al. Astigmatism-free Czerny-Turner compact spectrometer with cylindrical mirrors[J]. Applied Optics, 2017, 56(32): 9069–9073.
 Kingslake R. Who? Discovered coddington's Equations?[J]. Optics and Photonics News, 1994, 5(8): 20–23.
 Reader J. Optimizing Czerny–Turner spectrographs: a com-parison between analytic theory and ray tracing[J]. Journal of the Optical Society of America, 1969, 59(9): 1189–1196.
 Keough S J, Hanley T L, Wedding A B, et al. Grazing inci-dence X-ray studies of ultra-thin Lumogen films[J]. Surface Science, 2007, 601(24): 5744–5749.
 Wang L H, Wang X K, Chen B. Study of ultraviolet and extreme ultraviolet phosphors for imaging detector[J]. Optical Technique, 2006, 32(S1): 479–481.
王丽辉, 王孝坤, 陈波. 增强CCD紫外和极紫外成像的荧光物质的研究[J]. 光学技术, 2006, 32(S1): 479–481.
 Zhang D W, Tian X, Huang Y S, et al. Preparation and spectral characterization of Lumogen coatings for UV-responsive CCD image sensors[J]. Spectroscopy and Spectral Analysis, 2010, 30(5): 1171–1174.
张大伟, 田鑫, 黄元申, 等. CCD紫外敏感Lumogen薄膜制备与光谱表征[J]. 光谱学与光谱分析, 2010, 30(5): 1171–1174.
 Tao C X, Ruan J, Shu S P, et al. Thickness dependence of ultraviolet-excited photoluminescence efficiency of lumogen film coated on charge-coupled device[J]. Current Optics and Photonics, 2017, 1(4): 284–288.
 Du C G, Sun L Q, Ding Z T. Experiment study of enhancing CCD ultraviolet response using coronene[J]. Optical Tech-nique, 2010, 36(5): 753–757.
杜晨光, 孙利群, 丁志田. 利用晕苯增强CCD紫外响应的实验研究[J]. 光学技术, 2010, 36(5): 753–757.
 Zhao Y H, Yan F, Lou H W, et al. Measurement technology for relative spectral responsivity of the ultraviolet ICCD[J]. Spec-troscopy and Spectral Analysis, 2009, 29(5): 1371–1374.
 Zhao Y Q, Jiang L, He F, et al. Measurement and analysis of linear CCD nonlinear optical response characteristics[J]. Opto-Electronic Engineering, 2015, 42(7): 19–23.
赵友全, 江磊, 何峰, 等. 线阵CCD光电响应非线性特性测定与分析[J]. 光电工程, 2015, 42(7): 19–23.
 Mouroulis P, Wilson D W, Maker P D, et al. Convex grating types for concentric imaging spectrometers[J]. Applied Optics, 1998, 37(31): 7200–7208.
Open subject of Key Laboratory of General Optical Calibration and Characterization Technology of the Chinese Academy of Sciences (JZ2016QTXM1135)
Get Citation: Wang Guodong, Xia Guo, Li Zhiyuan, et al. Design and key technology research of portable UV-VIS spectrometer[J]. Opto-Electronic Engineering, 2018, 45(10): 180195.
Previous: The realizable method for large diameter liquid crystal optical phased array and the analysis of its far-field characteristics