The point measurement laser absorption spectroscopy (PMLAS) based on saturated absorption theory could surpass the defect of 'line-of-sight' measurement in traditional tunable diode laser absorption spectroscopy (TDLAS) and achieve the 'point' measurement with millimeter spatial resolution. It is realized by crossing with two frequency synchronized laser beams: one named probe beam as in traditional TDLAS and the other named saturated beam with higher power. In this paper, the theory of PMLAS was firstly analyzed by the theoretical deduction of saturated absorption coefficients with arbitrary cross angles and the numerical calculations of point absorbance under different saturation parameters. Next, a weak signal detection method based on high-frequency sinusoidal modulation of the saturated beam intensity was proposed, in which the first-order harmonic signal was theoretically deduced and verified by numerical demonstration. Furthermore, it is found that the FWHMs (full width at half maximum) of different order harmonics are all the same and equal to the width of the absorption signal without modulation, which implied that the superposition of multi-harmonics could enhance the signal-to-noise ratio (SNR) in measuring the spectrum line-width.
Theoretical research of point-measurement laser absorption spectroscopy
First published at:Oct 18, 2019
1 Wang S T, Wang C B, Pan Z, et al. Applications of optical technology in gas concentration detection[J]. Opto-Electronic Engineering, 2017, 44(9): 862-871. DOI:10.3969/j.issn.1003-501X.2017.09.002
王书涛, 王昌冰, 潘钊, 等.光学技术在气体浓度检测中的应用[J].光电工程, 2017, 44(9): 862-871. DOI:10.3969/j.issn.1003-501X.2017.09.002
2 Li L, Yang Y G, Chen W L, et al. Correction of influence conditions in TDLAS ammonia measuring based on hollow waveguide cell[J]. Opto-Electronic Engineering, 2015, 45(12): 35-40. DOI:10.3969/j.issn.1003-501X.2015.12.007
李龙, 杨燕罡, 陈文亮, 等.基于HWG气体池的TDLAS氨气测量中影响条件的修正[J].光电工程, 2015, 45(12): 35-40. DOI:10.3969/j.issn.1003-501X.2015.12.007
3 Sun P S, Zhang Z R, Xia H, et al. Study on real-time temperature measurement based on wavelength modulation technology[J]. Acta Optica Sinica, 2015, 35(2): 0230001.
4 Yin K W, Xu D, Zhang L, et al. 2D reconstruction for gas temperature and concentration based on TDLAS[J]. Opto-Electronic Engineering, 2016, 43(12): 20-27. DOI:10.3969/j.issn.1003-501X.2016.12.004
殷可为, 胥頔, 张龙, 等. TDLAS技术用于燃烧场气体温度和浓度重建研究[J].光电工程, 2016, 43(12): 20-27. DOI:10.3969/j.issn.1003-501X.2016.12.004
5 Jia L Q, Liu W Q, Kan R F, et al. Oxygen mass flow detection method in supersonic flow based on TDLAS[J]. Acta Photonica Sinica, 2015, 44(7): 0730001.
6 Hong G L, Zhang H P, Liu H, et al. Review of measurement for atmospheric CO2 differential absorption lidar[J]. Opto-Electronic Engineering, 2018, 45(1): 170452. DOI:10.12086/oee.2018.170452
洪光烈, 章桦萍, 刘豪, 等.国外差分吸收激光雷达探测大气CO2研究综述[J].光电工程, 2018, 45(1): 170452. DOI:10.12086/oee.2018.170452
7 Xu C, Fang Z H, Dong M L, et al. Design of non-invasive skin cholesterol detection system based on absorption spectroscopy[J]. Opto-Electronic Engineering, 2018, 45(4): 170587. DOI:10.12086/oee.2018.170587
许超, 方朝晖, 董美丽, 等.基于吸收光谱技术的皮肤胆固醇无创检测系统设计[J].光电工程, 2018, 45(4): 170587. DOI:10.12086/oee.2018.170587
8 Liu C, Xu L J, Cao Z, et al. One-dimensional tomography of axisymmetric temperature distribution with limited TDLAS data by using three-point Abel deconvolution[C]//Proceedings of 2014IEEE International Conference on Imaging Systems and Techniques, 2014: 432-435.
9 Xu L J, Liu C, Jing W Y, et al. Tunable diode laser absorption spectroscopy-based tomography system for on-line monitoring of two-dimensional distributions of temperature and H2O mole fraction[J]. Review of Scientific Instruments, 2016, 87(1): 013101. DOI:10.1063/1.4939052
10 Li J Y, Du Z H, Zhou T, et al. Non-uniform temperature distribution measurement of combustion field using temperature tuning multi-line thermometry[J]. Acta Optica Sinica, 2013, 33(S2): s212009.
李金义, 杜振辉, 周涛, 等.温度调谐多谱线方法测量燃烧场的非均匀温度分布[J].光学学报, 2013, 33(S2): s212009.
11 Nadir Z, Brown M S, Comer M L, et al. Gaussian mixture prior models for imaging of flow cross sections from sparse hyperspectral measurements[C]//Proceedings of 2015 IEEE Global Conference on Signal and Information Processing, 2015: 527-531.
12 Barone D L. Investigation of TDLAS measurements in a scramjet engine[D]. Cincinnati: University of Cincinnati, 2010.
13 Hannemann K. Hypersonic research in the high-enthalpy shock tunnel G ttingen[C]//Proceedings of the 30th International Symposium on Shock Waves 2, 2017: 1385-1390.
14 Ou D B, Chen L Z, Dong Y H, et al. Measurements of gas temperature and atomic oxygen density in the arc-heated wind tunnel based on TDLAS[J]. Journal of Experiments in Fluid Mechanics, 2015, 29(3): 62-67.
15 Kianvashrad N, Knight D. Effect of vibrational temperature boundary condition of isothermal wall on hypersonic shock wave laminar boundary layer interaction of a hollow cylinder flare[C]//Proceedings of the 7thEuropean Conference for Aeronautics and Aerospace Sciences, 2017.
16 Matsudo T, Takahara Y, Hori H, et al. Pseudomomentum transfer from evanescent waves to atoms measured by saturated absorption spectroscopy[J]. Optics Communications, 1998, 145(1-6): 64-68. DOI:10.1016/S0030-4018(97)00420-3
17 Jing Y F, Yan S B, Qin L, et al. Research of thin Rb vapor cell's saturated absorption spectroscopy[J]. Laser & Infrared, 2010, 40(7): 697-699. DOI:10.3969/j.issn.1001-5078.2010.07.004
荆彦锋, 闫树斌, 秦丽, 等.薄铷汽室饱和吸收光谱的研究[J].激光与红外, 2010, 40(7): 697-699. DOI:10.3969/j.issn.1001-5078.2010.07.004
18 Kumar P, Saini V K, Purbia G S, et al. Hyperfine structure studies of neutral europium transitions at 601.815 and 580.027 nm by saturation absorption spectroscopy[J]. Applied Optics, 2017, 56(6): 1579-1584. DOI:10.1364/AO.56.001579
19 Zizak G, Cignoli F, Benecchi S. Spatially resolved saturated absorption measurements of OH in methane-air flames[J]. Applied Optics, 1987, 26(19): 4293-4297. DOI:10.1364/AO.26.004293
20 Goldsmith J E M. Spatially resolved saturated absorption spectroscopy in flames[J]. Optics Letters, 1981, 6(11): 525-527. DOI:10.1364/OL.6.000525
21 Walters P E, Long G L, Winefordner J D. Spatially resolved concentration studies of ground state atoms and ions in an ICP: saturated absorption spectroscopic method[J]. Spectrochimica Acta Part B: Atomic Spectroscopy, 1984, 39(1): 69-76. DOI:10.1016/0584-8547(84)80007-5
22 Kychakoff G, Howe R D, Hanson R K. Spatially resolved combustion measurements using cross-beam saturated absorption spectroscopy[J]. Applied Optics, 1984, 23(9): 1303-1305. DOI:10.1364/AO.23.001303
23 Nomura S, Kaneko T, Komurasaki K. Development of highly sensitive and spatially resolved laser absorption spectroscopy for plasma wind tunnel measurement[C]//Proceedings of the 50thAIAA Aerospace Sciences Meeting Including the New Horizons Forum and Aerospace Exposition, 2012.
24 Nomura S, Ito G, Fujita K, et al. Translational temperature measurements in a shock layer by point-measurement laser absorption spectroscopy[C]//Proceedings of the 52ndAerospace Sciences Meeting, 2015.
25 Demtroder W. Laser Spectroscopy (Vol. 2): Experimental Techniques[M]. Ji Y, trans. Beijing: Science Press, 2012: 121-124.
26 Wang J, Huang W, Gu H T, et al. Gas temperature measurement with tunable diode laser absorption spectroscopy[J]. Acta Optica Sinica, 2007, 27(9): 1639-1642. DOI:10.3321/j.issn:0253-2239.2007.09.019
王健, 黄伟, 顾海涛, 等.可调谐二极管激光吸收光谱法测量气体温度[J].光学学报, 2007, 27(9): 1639-1642. DOI:10.3321/j.issn:0253-2239.2007.09.019
27 Kim S. Development of tunable diode laser absorption sensors for a large-scale arc-heated-plasma wind tunnel[D]. Stanford: Stanford University, 2004.
28 Parks T, McClellan J. Chebyshev approximation for nonrecursive digital filters with linear phase[J]. IEEE Transactions on Circuit Theory, 1972, 19(2): 189-194. DOI:10.1109/TCT.1972.1083419
Supported by National Natural Science Foundation of China (11702312)
Get Citation: Chen Wei, Wu Yue, Luo Jie, et al. Theoretical research of point-measurement laser absorption spectroscopy[J]. Opto-Electronic Engineering, 2019, 46(10): 180575.