Tunnel visibility has an important relationship with traffic safety. Real-time monitoring of tunnel visibility can provide a strong basis for ventilation and ventilation of tunnels. Based on the principle of visibility detection by transmission method, a set of visibility detection system is built for automatic detection of tunnel visibility. The detection system is mainly composed of a laser diode, a beam splitter, a silicon detector and a signal processing unit. Two silicon detectors detect the intensity of light before and after attenuation through optical path respectively and output corresponding photocurrent. The signal processing circuit amplifies and filters the output of the detector and converts the output to AD, and then calculates the extinction coefficient. Fixed transmittance filter is used to simulate tunnel environment, and the performance index of the detection system is tested in laboratory environment. The experimental results show that the measurement range of extinction coefficient is 3.37 km-1~118.82 km-1, the maximum relative measurement error is 8.4%, and the stability is better than 0.12 km-1.
Design of visibility detection system for tunnel based on transmission method
First published at:Oct 18, 2019
1 Dong L L, Shi N, Zhang L D, et al. Influence of tunnel lighting brightness on visibility under low transmittance[J]. Acta Photonica Sinica, 2017, 46(2): 0222002. DOI:10.3788/gzxb20174602.0222002
董丽丽, 石娜, 张利东, 等.低透过率下隧道照明亮度对能见度的影响[J].光子学报, 2017, 46(2): 0222002. DOI:10.3788/gzxb20174602.0222002
2 Ju L. Hardware design of transmission visibility measurement based on WSN[D]. Nanjing: Nanjing University of Information Science and Technology, 2011: 5–7.
3 Gao Q. Development of the infrared forward scattering visibility meter[D]. Tianjin: Tianjin University, 2015: 1–3.
4 Wang J A, Kang S, Wu R H, et al. Method for calculating visibility utilizing optic ways[J]. Acta Optica Sinia, 2009, 29(1): 283–289. DOI:10.3772/j.issn.1009-5659.2010.23.013
王江安, 康圣, 吴荣华, 等.能见度光学测量方法[J].光学学报, 2009, 29(1): 283–289. DOI:10.3772/j.issn.1009-5659.2010.23.013
5 Cai Q. Analysis of performance for transmission type and forward scatter type apparatus[J]. Journal of Chengdu Technological University, 2014, 17(4): 40–42, 59. DOI:10.13542/j.cnki.51-1747/tn.2014.04.013
蔡琴.透射式和前散射式能见度仪性能的分析[J].成都工业学院学报, 2014, 17(4): 40–42, 59. DOI:10.13542/j.cnki.51-1747/tn.2014.04.013
6 Cheng S R, Wei Q Z, Lv J. The development of a new atmosphere transmittance meter[J]. Opto-Electronic Engineering, 2011, 38(2): 144–150. DOI:10.3969/j.issn.1003-501X.2011.02.024
程绍荣, 魏全忠, 吕军.一种实用型大气透射式能见度仪的研制[J].光电工程, 2011, 38(2): 144–150. DOI:10.3969/j.issn.1003-501X.2011.02.024
7 Wang Z L, Cao N F, Wang C L. New design of laser instrument for detecting the visibility[J]. Laser & Infrared, 2012, 42(6): 629–632. DOI:10.3969/j.issn.1001-5078.2012.06.007
王宗俐, 曹乃锋, 王春录.一种激光能见度仪设计的新方法[J].激光与红外, 2012, 42(6): 629–632. DOI:10.3969/j.issn.1001-5078.2012.06.007
8 Gong C W, Li X B, Li J Y, et al. New method of aerosol extinction coefficient measurement[J]. Acta Optica Sinica, 2014, 34(1): 0101001. DOI:10.3788/AOS201434.0101001
宫纯文, 李学彬, 李建玉, 等.大气气溶胶消光系数测量新方法[J].光学学报, 2014, 34(1): 0101001. DOI:10.3788/AOS201434.0101001
9 Zhuang Z B, Tai H D, Jiang L H. Changing baseline lengths method of visibility measurement and evaluation[J]. Acta Optica Sinia, 2016, 36(2): 0201001. DOI:10.3788/AOS201636.0201001
庄子波, 台宏达, 蒋立辉.一种基线长度改变的能见度测量和评价方法[J].光学学报, 2016, 36(2): 0201001. DOI:10.3788/AOS201636.0201001
10 Ministry of Transport of the People's Republic of China. Specifications for design of highway tunnels section 2 traffic engineering and affiliated facilities: JTG D70/2–2014[S]. Beijing: China Communications Press, 2014.
11 Bendix J. Determination of fog horizontal visibility by means of NOAA－AVHR[C]//1995 International Geoscience and Remote Sensing Symposium, IGARSS '95. Quantitative Remote Sensing for Science and Applications, Firenze, Italy, 1995, 3: 1847–1849.
12 Wu J, Yang C P, Liu J B. Optical Transmission Theory in the Atmosphere[M]. Beijing: Beijing University of Posts and Telecommunications Press, 2005: 1–57.
13 Du D Z, Liu T T, Liao W H, et al. Design of monitoring system of melt pool light intensity in selective laser melting[J]. Infrared and Laser Engineering, 2017, 46(12): 1206002. DOI:10.3788/IRLA201746.1206002
杜道中, 刘婷婷, 廖文和, 等.激光选区熔化熔池光强监测系统设计[J].红外与激光工程, 2017, 46(12): 1206002. DOI:10.3788/IRLA201746.1206002
14 Zhu Z. Design and stability analysis of pre-amplification circuit for photodiode monitoring[J]. Modern Electronics Technique, 2014, 37(17): 137–139. DOI:10.3969/j.issn.1004-373X.2014.17.041
朱珠.光电检测前置放大电路设计及稳定性分析[J].现代电子技术, 2014, 37(17): 137–139. DOI:10.3969/j.issn.1004-373X.2014.17.041
15 Gao L M. Measurement of CO concentration and visibility[D]. Hangzhou: Zhejiang University, 2013: 55–56.
Supported by National Natural Science Foundation of China (11703061) and National Key R & D Program of China(2016YFB0500704)
Get Citation: Ye Song, Bai Yunfei, Li Zhiwei, et al. Design of visibility detection system for tunnel based on transmission method[J]. Opto-Electronic Engineering, 2019, 46(10): 180607.